Atmospheric Correction and Vicarious Calibration of Oceansat-1 Ocean Color Monitor (OCM) Data in Coastal Case 2 Waters
NAVAL RESEARCH LAB STENNIS DETACHMENT STENNIS SPACE CENTER MS OCEANOGRAPHY DIV
Pagination or Media Count:
The Ocean Color Monitor OCM provides radiance measurements in eight visible and near-infrared bands, similar to the Sea-viewing Wide Field-of-View Sensor SeaWiFS but with higher spatial resolution. For small- to moderate-sized coastal lakes and estuaries, where the 1 x 1 km spatial resolution of SeaWiFS is inadequate, the OCM provides a good alternative because of its higher spatial resolution 240 x 360 m and an exact repeat coverage of every two days. This paper describes a detailed step-by-step atmospheric correction procedure for OCM data applicable to coastal Case 2 waters. This development was necessary as accurate results could not be obtained for our Case 2 water study area in coastal Louisiana with OCM data by using existing atmospheric correction software packages. In addition, since OCM-retrieved radiances were abnormally low in the blue wavelength region, a vicarious calibration procedure was developed. The results of our combined vicarious calibration and atmospheric correction procedure for OCM data were compared with the results from the SeaWiFS Data Analysis System SeaDAS software package outputs for SeaWiFS and OCM data. For Case 1 waters, our results matched closely with SeaDAS results. For Case 2 waters, our results demonstrated closure with in situ radiometric measurements, while SeaDAS produced negative normalized water leaving radiance nLw and remote sensing reflectance Rrs. In summary, our procedure resulted in valid nLw and Rrs values for Case 2 waters using OCM data, providing a reliable method for retrieving useful nLw and Rrs values which can be used to develop ocean color algorithms for in-water substances e.g., pigments, suspended sediments, chromophoric dissolved organic matter, etc. at relatively high spatial resolution in regions where other software packages and sensors such as SeaWiFS and Moderate Resolution Imaging Spectrometer MODIS have proven unsuccessful. The method described here can be applied to other sensors.
- Hydrology, Limnology and Potamology