Package Integrity Measurement Technology and Quality Assurance
Final rept. Oct 1989-Mar 1992
RAYTHEON CO SUDBURY MA EQUIPMENT DIV
Pagination or Media Count:
This report is the result of a study to evaluate two procedures in MIL-STD-883D, Test Method 1014.9 Seal and Test method 1018.2 Internal Water- Vapor Content. The first portion of this report reviews and investigates the current test procedures included in Test Method 1014 and explores new test methods for recommendation of incorporation into a new revision of the test method. Included in Appendix-B are the results of a survey of industry for comments and recommendations for improving and clarifying the present test method. Appendix A of this report presents the proposed new version of Test Method 1014. The second portion of this report details the results of a study that provided and distributed Test Method 1018 moisture measurement correlation samples to various Residual Gas Analysis RGA laboratories. This was done in order to determine the accuracy and precision of mass spectrometric measurement of moisture in microelectronics packages at facilities. deemed suitable or candidates to be deemed suitable by the Defense Electronic Supply Center DESC to perform MIL-STD-883D, Test Method 1018.2, Procedure 1 Mass Spectrometry, moisture testing. Hermeticity, RGA, Laa correlation, Moisture, Test method 1018 Internal Water-Vapor Content, Test method 1014.
- Containers and Packaging
- Test Facilities, Equipment and Methods